Page:Radio-activity.djvu/117

 '''63. Electrometer key.''' For work with electrometers of high sensibility, a special key is necessary to make and break from a distance the connection of the quadrants with earth in order to avoid electrostatic disturbances at the moment the current is to be measured. The simple key shown in Fig. 16 has been found very satisfactory for this purpose. A small brass rod BM, to which a string is attached, can be moved vertically up and down in a brass tube A, which is rigidly attached to a bent metal support connected with earth. When the string is released, this rod makes contact with the mercury M, which is placed in a small metal vessel resting on a block of ebonite P. The electrometer and testing vessel are connected with the mercury. When the string is pulled, the rod BM is removed from the mercury and the earth connection of the electrometer system is broken. On release of the string, the rod BM falls and the electrometer is again earthed. By means of this key, which may be operated at any distance from the electrometer, the earth connection may be made and broken at definite intervals without any appreciable disturbance of the needle. Fig. 16. 64. Testing apparatus. The arrangement shown in Fig. 17 is very convenient for many measurements in radio-activity. Two parallel insulated metal plates A and B are placed inside a metal vessel V, provided with a side door. The plate A is connected with one terminal of a battery of small storage cells, the other pole of which is earthed; the plate B with the electrometer, and the vessel V with earth. The shaded areas in the figure indicate the position of ebonite insulators. The active material to be tested is spread uniformly in a shallow groove (about 5 cms. square and 2 mms. deep) in the brass plate A. In order to avoid breaking the battery connection every time the plate A is removed, the wire from the battery is permanently connected with the metal block N