Page:Micrographia - or some physiological descriptions of minute bodies made by magnifying glasses with observations and inquiries thereupon.djvu/89

Rh. This vessel being thus prepared, the way of marking and graduating the Thermometers may be very easily thus performed:

Fill this Cylindrical vessel with the same liquor wherewith the Thermometers are fill'd, then place both it and the Thermometer you are to graduate, in water that is ready to be frozen, and bring the surface of the liquor in the Thermometer to the first marke or [0]; then so proportion the liquor in the Cylindrical vessel, that the surface of it may just be at the lower end of the small glass-Cylinder; then very gently and gradually warm the water in which both the Thermometer and this Cylindrical vessel stand, and as you perceive the ting'd liquor to rise in both stems, with the point of a Diamond give several marks on the stem of the Thermometer at those places, which by comparing the expansion in both Stems, are found to correspond to the divisions of the cylindrical vessel, and having by this means marked some few of these divisions on the Stem, it will be very easie by these to mark all the rest of the Stem, and accordingly to assign to every division a proper character.

A Thermometer, thus marked and prepared, will be the fittest Instrument to make a Standard of heat and cold that can be imagined. For being sealed up, it is not at all subject to variation or wasting, nor is it liable to be changed by the varying pressure of the Air, which all other kind of Thermometers that are open to the Air are liable to. But to proceed.

This property of Expansion with Heat, and Contraction with Cold, is not peculiar to Liquors only, but to all kind of solid Bodies also, especially Metals, which will more manifestly appear by this Experiment.

Take the Barrel of a Stopcock of Brass, and let the Key, which is well fitted to it, be riveted into it, so that it may slip, and be easily turned round, then heat this Cock in the fire, and you will find the Key so swollen, that you will not be able to turn it round in the Barrel; but if it be suffered to cool again, as soon as it is cold it will be as movable, and as easie to be turned as before.

This Quality is also very observable in Lead, Tin, Silver, Antimony, Pitch, Rosin, Bees-wax, Butter, and the like; all which, if after they be melted you suffer gently to cool, you shall find the parts of the upper Surface to subside and fall inwards, losing that plumpness and smoothness it had whilst in fusion. The like I have also observed in the cooling of Glass of Antimony, which does very neer approach the nature of Glass,

But because these are all Examples taken from other materials then Glass, and argue only, that possibly there may be the like property also in Glass, not that really there is; we shall by three or four Experiments indeavour to manifest that also.

And the First is an Observation that is very obvious even in these very drops, to wit, that they are all of them terminated with an unequal or irregular Surface, especially about the smaller part of the drop, and the whole length of the stem; as about D, and from thence to A, the whole Surface, which would have been round if the drop had cool'd leisurely, is, by being quenched hastily, very irregularly flatted and pitted; which Rh